<?xml version="1.0" encoding="ISO-8859-1"?>

<FDSNStationXML xmlns:xsi="http://www.w3.org/2001/XMLSchema-instance" xmlns="http://www.fdsn.org/xml/station/1" schemaVersion="1.0" xsi:schemaLocation="http://www.fdsn.org/xml/station/1 http://www.fdsn.org/xml/station/fdsn-station-1.0.xsd">
 <Source>IRIS-DMC</Source>
 <Sender>IRIS-DMC</Sender>
 <Module>IRIS WEB SERVICE: fdsnws-station | version: 1.0.11</Module>
 <ModuleURI>https://service.iris.edu/fdsnws/station/1/query?channel=LKS&amp;station=CMB&amp;network=BK&amp;starttime=2004-06-15T00%3A00%3A00.000000&amp;level=response</ModuleURI>
 <Created>2014-02-16T15:50:00</Created>
 <Network code="BK" startDate="1980-01-01T00:00:00" endDate="2500-12-12T23:59:59" restrictedStatus="open">
  <Description>Berkeley Digital Seismograph Network</Description>
  <TotalNumberStations>80</TotalNumberStations>
  <SelectedNumberStations>1</SelectedNumberStations>
  <Station code="CMB" startDate="1996-09-25T19:19:00" endDate="2599-12-31T23:59:59" restrictedStatus="open" alternateNetworkCodes=".EARTHSCOPE,_GSN-BROADBAND,_US-TA,_GSN,_FDSN,_US-REGIONAL,.UNRESTRICTED,_ANSS,_US-REF,_US-ALL,_ANSS-BB,.EARTHSCOPE,_FDSN-ALL,_US-ALL,_REALTIME,_STS-1">
   <Latitude>38.03455</Latitude>
   <Longitude>-120.38651</Longitude>
   <Elevation>697.0</Elevation>
   <Site>
    <Name>Columbia College, Columbia, CA, USA</Name>
   </Site>
   <CreationDate>1986-10-25T00:00:00</CreationDate>
   <TotalNumberChannels>111</TotalNumberChannels>
   <SelectedNumberChannels>2</SelectedNumberChannels>
   <Channel locationCode="  " startDate="2004-06-15T00:00:00" restrictedStatus="open" endDate="2010-12-17T00:00:00" code="LKS">
    <Latitude>38.03455</Latitude>
    <Longitude>-120.38651</Longitude>
    <Elevation>697.0</Elevation>
    <Depth>2.0</Depth>
    <Azimuth>0.0</Azimuth>
    <Dip>0.0</Dip>
    <Type>CONTINUOUS</Type>
    <Type>GEOPHYSICAL</Type>
    <SampleRate>1.0</SampleRate>
    <ClockDrift>1.0E-4</ClockDrift>
    <CalibrationUnits>
     <Name>A</Name>
     <Description>Amperes</Description>
    </CalibrationUnits>
    <Sensor>
     <Type>YSI 44031 Thermistor</Type>
    </Sensor>
    <Response>
     <InstrumentSensitivity>
      <InputUnits>
       <Name>C</Name>
       <Description>Degrees Centigrade</Description>
      </InputUnits>
      <OutputUnits>
       <Name>COUNTS</Name>
       <Description>Digital Counts</Description>
      </OutputUnits>
     </InstrumentSensitivity>
     <Stage number="1">
      <Polynomial>
       <InputUnits>
        <Name>C</Name>
        <Description>Degrees Centigrade</Description>
       </InputUnits>
       <OutputUnits>
        <Name>V</Name>
        <Description>Volts</Description>
       </OutputUnits>
       <ApproximationType>MACLAURIN</ApproximationType>
       <FrequencyLowerBound>0.0</FrequencyLowerBound>
       <FrequencyUpperBound>0.016</FrequencyUpperBound>
       <ApproximationLowerBound>3.4</ApproximationLowerBound>
       <ApproximationUpperBound>68</ApproximationUpperBound>
       <MaximumError>.1</MaximumError>
       <Coefficient number="2">12.5091</Coefficient>
       <Coefficient number="3">13.8687</Coefficient>
       <Coefficient number="4">4.31802</Coefficient>
       <Coefficient number="5">1.44163</Coefficient>
       <Coefficient number="6">.829113</Coefficient>
       <Coefficient number="7">.738976</Coefficient>
       <Coefficient number="8">.41094</Coefficient>
       <Coefficient number="9">.121068</Coefficient>
       <Coefficient number="10">.017856</Coefficient>
       <Coefficient number="11">.001043</Coefficient>
      </Polynomial>
     </Stage>
    </Response>
   </Channel>
  </Station>
 </Network>
</FDSNStationXML>